Please consider our TechNote of the sterilization methods of the mp6 micropumps. For ebeam and gamma sterilization we had a customer project where a system with the pump was tested before and after the sterilization processes. No impact of the processes could be detected, except for a slight change of the colour. Functionality and flow rate were unchanged. For both methods the radiatons of 15, 25 and 40 kGy were experimented successfully.
Kategorien: mp6, Quality