Please find here an example for lower flow rates which were generated with the mp-Lowdriver and the mp6 micropump.
The mp6-series micropumps always work corresponding to the pump curve, which is based on the maximum flowrate (at zero pressure) and the maximum pressure (at zero flowrate). The curve of pressure and flowrate combinations that are directly achievable with the … Weiter
Pumps in series connection can be used in order to create underpressures on the inlet side. The underpressure that can be reached follows an exponential curve with multiple pumps, at a certain point further pumps cannot build up further underpressure. … Weiter
A Pressure reservoir could be set to contain overpressure or vacuum. A valve would then be the required precise tool letting the vacuum or overpressure gain access to the test tubing. Two 2/2 valves or one 3/2 valve would be used … Weiter
All wetted materials inside the standard mp6 are PPSU.. Please note that the mp6 is made with material of two grades of PPSU; natural-transparent and black. The lid is black. Radel R-5000 NT is the natural-transparent material and Radel R-5100 … Weiter
Liquids: The maximum fluidic power is generated by a pump when operating it at half of the maximum flow rate and half of the maximum back pressure. Therefore, these operating conditions will be considered in the following: Operating the mp6 … Weiter
Due to the functional principle of the piezoelectric diaphragm pumps a pulsation free flow can’t be generated with just the pump. We would recommend to use a damper, for example our developed mp-damper. However, as the displaced volumes are very … Weiter
Please consider our TechNote of the sterilization methods of the mp6 micropumps. For ebeam and gamma sterilization we had a customer project where a system with the pump was tested before and after the sterilization processes. No impact of the … Weiter